Conductivity Controller 2 SetPoint
Work Instruction Digital Electrical Conductivity Controller For Dosing; Conductivity Controller 2 SetPoint
1. SETTING RELAY OPERATION
SET High limit and Low limit of Electrical Conductivity Value for Buzzer / Relay Operation
a) Press the SET button Upper display shows H-EC the higher limit of Electrical Conductivity to set and lower display shows the value of H-EC already set. Now use / Press UP and DN buttons to set required Higher Limit
b) Press SET button once again Upper display shows L-EC the lower limit of Conductivity to set and lower display shows the value of L-EC already set Now use / Press UP and DN buttons to set required Lower Limit of Conductivity
c) Press SET Button again or ENT button to save the changes else change will not save in memory and remove after power OFF
Automatic Conductivity Controller ; Conductivity Controller 2 SetPoint
While PPM (Particle per Million) TDS value (Total Dissolved Solids) gives the sum of total dissolved solids in water. Hence These solids may be salts, minerals, and metals. Mostly Sodium and magnesium salt fond dissolved. TDS measred in PPM.
Hence The conductivity of the water or any solid / liquid measured; indicates ions are in the water, the better it conducts electricity. While TDS meters typically quantify this conductivity; in micro siemens UCM or MCM. Hence The latter stands for parts per million; i.e. the number of solid particles per one million water mixture particles. A value of 40 ppm means that from one million particles there are 40 dissolved ions and the rest (= 999 960) are water molecules. TDS Controller 2 SetPoint
Best Conductivity Controller On India
What does it mean about the water quality?
Hence DM water with a ppm value of zero, is automatically free from any pesticides and drug residues, is far from it. While All the TDS meter measures are merely the number of ions in the water. Most of the pesticides; hormone-disturbing substances, and drug residues are usually no ions.
Digital Conductivity Controller 2 SetPoint
Hence In short: water with a low ppm value is virtually free from salts and minerals; but can hide a large cocktail of non-ionic chemicals in itself. While The result of a TDS meter is therefore not very meaningful; with regard to the actual water quality. while Other analytical methods or a water analysis by a reputable laboratory; may be better suited to find out about the actual water quality.
Observation as analyst TDS Controller 2 SetPoint
Hence if you want to check the functionality of an osmosis system; Then the mere TDS values are not suitable as an indicator of the water quality. Hence extensive and detailed water analyzes are often very expensive and only reflect a snapshot; While they are only advisable if the drinking water is to be tested; for individual substances. Hence It often makes more sense to invest; in a high-quality water filter right away. While This way you have more security with your daily drinking water; consumption in the long term. Hence it is essential to pay attention to appropriate assessments and long-term studies.
- PRODUCT SPECIFICATION :
- Range : 0000 to 1999 ucm /9999 ucm
- Resolution : 1 ucm
- Tolerance : +/- 1%
- Set-Point: 2 Nos Lower level and Higher Level
- Output : 2 Relay Output
- Calibration: Knobe Provided
- Input: Conductivity Sensor
- Display: Seven-Segment Bright LED Display
- Operating Voltage: 220 Vac
- Size : 96 * 96 * 110
- Body: PP/ABS
Based on 0 reviews